The basic d.c. voltmeter is nothing but a PMMC D'Arsonoval galvanometer. The resistance is required to be connected in series with the basic meter to use it as a voltmeter.

This series resistance is called a multiplier. The main function of the multiplier is to limit the current through the basic meter so that the meter current does not exceed the full scale deflection value. The voltmeter measures the voltage across the two points of a circuit or a voltage across circuit component. The basic d.c. voltmeter is shown in the Fig. 1.

Fig. 1 Basic of d.c. voltmeter |

The voltmeter must be connected across the two points or a component, to measure the potential difference, with the proper polarity.

The multiplier resistance can be calculated as:

Let R

_{m }= Internal resistance of coil i.e. meter
R

_{s }= series multiplier resistance
I

_{m }= full scale deflection current
V = full range voltage to be measured

From Fig 1,

**.**V = I^{.}._{m }(R_{m }+ R_{s })**.**V = I

^{.}._{m}R

_{m }+ I

_{m }R

_{s }

The multiplying factor for multiplier is the ratio of full range voltage to be measured and the drop across the basic meter.

Let v = drop across the basic meter = I

_{m}R_{m }
Hence multiplier resistance can also be expressed as,

R

_{s }= (m-1)
Thanks for sharing details of electrical panels. I stumbled here to learn more about electrical panels because my friend's firm is one of the leading electrical panel manufacturers in India

ReplyDeleteA voltmeter is an instrument that is used to check and measure the amount of voltage that is passing between two points in an electric current. Omega Fuses

ReplyDelete